After years of working with new thermographers in our training classes, I’ve learned a lot about how people learn. First, we never get it right the first time! Don’t worry about making mistakes, in fact, it is important to make them. But it is also equally important to learn from them. In class we let [...]
Last week we discussed how to adjust level and span manually. Sometimes this can lead to a lot of “button pushing!” It is not a big deal – just one of life’s little aggravations. Thankfully, many models of Fluke imagers have a very helpful solution: 1-TIME AUTO ADJUST.
Both images were taken using AUTO adjust. [...]
Last week we discussed how to adjust level and span automatically and when that feature is best used. While using AUTO can be very convenient in many situations, you’ll quickly find times when MANUAL adjustment is essential, either to getting the best image or understanding what you are seeing.
Use AUTO adjust to get “in [...]
I’m all for using the AUTO adjust feature found on most thermal imagers. That said, it does not mean I can turn my brain off! Many new thermographers believe they can just put their system on AUTO and life will be good. Not true.
AUTO adjusts the span of the image to accommodate all three [...]
The image palettes we have available to us are remarkable. They’ve come so far since I first got into this business in 1983. We can now also easily change the palette in the software. Wow!
The Fluke AMBER palette with red and blue saturation indicators is a nearly perfect palette to use, especially in the [...]
When Fluke IR-fusion® was first launched, I wondered whether or not it would be perceived as just another marketing gimmick, and if not, how well it would be used. Time has proven that few find it gimmicky and most use it very well.
In the field I nearly always use the Full Infrared setting (left) for [...]
We’ve spent the past several weeks discussing heat transfer. Radiation is a special mode of transfer because that is the mode by which energy is transferred from surfaces to our imagers.
All of what we see in the thermal image is nearly always based on the amount of radiation emitted from or reflected by the surface. [...]
For young children and pets, the first look in a mirror can be very confusing! They may be asking themselves. “Is that real or…?” Most thermographers share a similar exasperation the first time they see a thermal reflection, most commonly from a bright piece of metal. Many go on to understand that reflections are not [...]
For the past several weeks, we’ve been reviewing heat transfer. Thermographers must understand the basics if they are to successfully interpret their images. Over the next two weeks, we’ll wrap up the review with a discussion of radiation.
Electromagnetic radiation is not only a powerful mode of heat transfer, it is also the way energy moves [...]
As we continue the discussion about convective heat transfer, it is useful to define two types of convection, natural and forced. When quantities of fluids are moved in either way, we also use the term “bulk energy transfer” because it is really the movement of the fluid itself, and the energy inherent in it, that [...]